Payment & Shipping Terms:
|Work Environment:||0~40℃, Humidity:≤85%RH||Voltage Grade (kV):||AC/DC50kV|
high voltage test kit,
electronic test equipment
FRC series AC / DC high voltage voltage divider is a resistance-capacitance equipotential shielding voltage divider type high voltage measuring device. It is mainly used for the measurement of pulse high voltage, lightning high voltage, and power frequency high voltage. Is the preferred alternative to high-voltage electrostatic voltmeter.
High Precision High Voltage Divider has the characteristics of simple operation, intuitive display, high accuracy, small size, light weight, etc. It is suitable for power plants, substations, high-voltage electrical equipment manufacturing plants, high-voltage laboratory and other departments as ideal equipment for high-voltage measurement.
FRC AC/DC HV Divider is on-site instrument, which test both AC and DC voltage.
It consist of voltage divider and LCD display measurement meter
1.Balance type equal potential shield structure.
2.Fully-sealed insulation cylinder with internal excellent components.
3.High accuracy and linearity, good stability and anti-interference ability.
4.LCD screen to show AC RMS, peak/√2peak factor, average value, peak-to-peak value, voltage waveform and frequency.
5.Inner shield anti-interference system.
6.The voltage ratio can be customized, suitable for calibration.
Technique parameter :
|Voltage Grade (kV)||AC/DC50kV||AC/DC100kV||AC/DC150kV||AC/DC300kV|
|Voltage division ratio||1000:1||1000:1||1000:1||1000:1|
|Work Environment||0~40℃, humidity:≤85%RH|
Work Environment: temperature:0~40℃, humidity:≤85%RH
Preparation Before Test (Measurement):
(1)High voltage divider is arranged in a proper position. Be sure that, an adequate clearance should be kept to surrounding objects. The minimum clearance should not be less than the height of the voltage divider.
(2)Grounding connection: The ground terminals of voltage divider basement and of digital voltmeter should be reliably connected to ground by braided copper wire.
(3)Measuring signal connection: The supplied coaxial measuring cable is specially prepared for connecting from voltage signal output socket of divider bacement to the signal input socket of digital voltmeter. If should not be replaced by other coaxial cable or any connecting wire, because of capacitive match to the voltage divider.
(4)High voltage connection: The high voltage terminal to be measured is connected to the top connecting screw MIO of voltage divider by HV leads. Be sure that, an adequate angle of 75°~90°between HV leads (inclined downward) and vertical axis of voltage divider should be kept during calibration and use for precise measurement.
Wuhan GDZX Power Equipment Co.,Ltd OFFER YOU :
1: FACTORY DIRECT PRICE.
2: NICE QUALITY.
3: QUICK SHIPPING WITHIN 15 DAYS.
4: SMALL ORDER WELCOME..
5: ONLINE PAYMENT.
6: SUPPORT One year free warranty, three years quality assurance, lifetime maintenance.
7: AFTER SALES SERVICES 24/7
Our Sevice :
Signing Contracts And Technical Agreements
Ensure Execution Of Contract And Agreement
Complete After-Sales Service
Professional After-Sale Service Team
1 Year Free Warranty, Life-Long Maintenance
Help Troubleshooting Within 24 Hours
Professional R & D Team
Contact Person: email@example.com